January 23, 2008
Thank you for choosing Nikon Microscopes.
Nikon has recently discovered that with certain units of biological microscopes Eclipse E100 and E200 series,the lamp output voltage may not rise to rating.
The light intensity is enough for bright field observation, but may not be enough for phase contrast observation. If you feel the illumination is not bright enough, please contact your nearest Nikon representative or customer service center shown below.
Even if the product is already out of warranty, Nikon will offer a free replacement of parts.
Please note that this defect dose not directly lead to decrease of product safety.
Nikon regrets any inconvenience this matter may cause and we appreciate your continued support of our products.
Products subject to replacement:
Certain units of The E100, E200 and E200-Pol shipped from Nikon Corporation from January to November 2007.
For contacting Nikon
In the Americas
Nikon Instruments INC.
1300 Walt Whitman Road, Melville, NY 11747-3064, U.S.A.
Tel: +1-631-547-8500 Fax: +1-631-547-0306
Nikon Instruments Europe B.V.
Laan van Kronenburg 2,1183 As Amstelveen, The Netherlands
Tel: +31-20-44-96-222 Fax: +31-20-44-96-298
Nikon Instruments (Shanghai) Co., Ltd.
26F, Hang Seng Bank Tower, No.1000 Lujiazui Ring Road,
Pudong New District, Shanghai 200120, China
Beijing Branch TEL: +86-10-5869-2255 FAX: +86-10-5869-2277
Guangzhou Branch TEL: +86-20-3882-0552 FAX: +86-20-3882-0580
Nikon Instruments Korea Co., Ltd.
1F, Yuk-In B/D, 996-16, Daechi-2Dong,
Kangnam-Ku, Seoul, South Korea 135-502
TEL: +82-2-2186-8400 FAX: +82-2-555-4415
In Singapore, Malaysia, Thailand, Indonesia, Vietnam, Philippines, India
Nikon Singapore Pte. Ltd.
80 Anson Road, #10-01/02
Fuji Xerox Towers
Singapore 079907, Singapore
TEL: +65-6559-3618 FAX: +65-6559-3668
Other Asian Countries, Middle-East, Oceania, etc.
Marketing Department, Microscope Solutions Business Unit
Shinagawa Intercity Tower C, 2-15-3, Konan, Minato-ku, Tokyo 108-6290 Japan
Tel: +81-3-6433-3705 FAX:+81-3-6433-3785
Bio Science CS Engineering Section Quality Assurance Department, Instruments Company
Tel: +81-45-853-8525 9:00am - 5:00pm weekdays (JST)
For out-of-hours support, contact us by e-mail. e-mail: firstname.lastname@example.org
- *Organization name of Nikon Corporation was revised on May 22, 2009.
- *Sales Dept., Bio Science Division, Industrial Instruments Company moved office on May 6, 2010.
- *Nikon Instruments (Shanghai) Co., Ltd. moved office on July 12, 2010.